diasend® Clinic – Uploading Roche Accu-Chek Spirit Combo insulin pump
Hi! Welcome to diasend®. My name is Kristy. In this video we will show you how to upload the Accu-Chek insulin pump. This is the Roche Accu-Chek Spirit Combo. To upload it connect the Smartpix to the Transmitter… …and wait for the blue indicator on the Smartpix to start flashing slowly. Then put the pump in “PC link” mode. To do this, stop the pump and enter the “PC connection” menu. Then place the pump with its infrared eye, that is here, towards the Smartpix. The blue indicator signals that the data is being transmitted to the Smartpix. Wait for the blue indicator to start flashing slowly again, or stop flashing altogether. You will see that the blue light would change from flashing intermittently to flashing quickly. The data is now being transmitted from the Smartpix to the Transmitter. Do not disconnect the Smartpix yet. Wait for the Transmitter to signal that the upload was successful. When the green light comes on the device is uploaded and you can now view the data in diasend®. You should now disconnect the Smartpix from the transmitter and connect the cable for the USB hub. We hope this video was helpful! If you have any questions please visit us at www.diasend.com or contact our customer support team.